Fluke 5618B-9-X Small Diameter Industrial RTD
For secondary level performance with full ITS-90 calibration, Fluke Calibration’s 5618B series PRTs are an excellent choice for critical temperature measurements.
- done HTTPS SECURE: Our pages are encrypted and protected from cyber-attack.
- done COMPLIANT: Payment Card Industry Data Security Standard (PCI DSS) Compliant.
- done PROTECTED: by TLS (SSL) - all data/information is secure from interception.
Featuring a 3.2 mm diameter (1/8 in) sheath, these industrial standards probes have reduced response time without compromising precision. This small diameter 5618A probe works well in many applications where immersion depth is limited. Larger diameter probes give more measurement error in short immersion depth applications because they conduct more heat between ambient and the sensor.
The Fluke 5618B is also a great probe to use for calibrating your Fluke Calibration 9132 or 9133 infrared calibrators. In fact, these IR black body heat sources were designed to be calibrated with this type of probe. We recommend using the Fluke 5618B PRTs with the 1523, 1524, 1502A, 1529, or 1560 thermometer readouts.
Please Note: When ordering please specify the termination you require. ie. Fluke 5618B-9-G (gold pins).
X = termination.
Specify “A" (INFO-CON for 914X), “B" (bare wire), “D" (5-pin DIN for Tweener Thermometers), “G" (gold pins), “I" (INFO-CON for 1521 or 1522), “J" (banana plugs), “L" (mini spade lugs), "M" (mini banana plugs), “P" (INFO-CON for 1523 or 1524), or “S" (spade lugs).
229 mm (9 in) Small Diameter Probe, lead wire (cable) 6 ft.
Small diameter sheath, 3.2mm (0.125 in).
Includes ITS-90 coefficients.
NVLAP-accredited calibrated from –200°C to 500°C, lab code 200706-0.
|Resistance||Nominal 100 Ω at 0 °C.|
|0.003923 Ω/Ω/ °C nominal.|
|–200 °C to 500 °C
(–200 °C to 300 °C for 5618B-6-X).
|Drift Rate||± 0.1 °C when used periodically to 500 °C.|
|Sheath Material||316 SST.|
|Leads||22 AWG PTFE, 6’.|
||Less than 0.01 °C at 0 °C when using –196 °C and 420 °C as the end points.|
||Four seconds maximum for 63.2 % response to step change in water moving at 3 fps.|
|Thermal EMF||Less than 25 mV at 420 °C.|
||Includes manufacturer’s NVLAP-accredited calibration w/ITS-90 coefficients, R vs. T values in 1 °C increments, lab code 200706-0.|
||5618B-12: 305 mm L x 3.2 mm dia. (12 x 1/8 in)
5618B-9: 229 mm L x 3.2 mm dia. (9 x 1/8 in)
5618B-6: 152 mm L x 3.2 mm dia. (6 x 1/8 in)
|± 0.026 °C at –200 °C
± 0.066 °C at 0 °C
± 0.219 °C at 500 °C.
No customer reviews for the moment.