Fluke 9103 Field Dry-Well
Includes: 9103 Dry-well, Insert for configuration ordered, Power Cord, Insert Removal Tongs, 9930 Software, Serial Cable, Manual.
- done HTTPS SECURE: Our pages are encrypted and protected from cyber-attack.
- done COMPLIANT: Payment Card Industry Data Security Standard (PCI DSS) Compliant.
- done PROTECTED: by TLS (SSL) - all data/information is secure from interception.
The Fluke 9103 Dry-well may be used as a portable instrument or bench top temperature calibrator for calibration thermocouples and RTD temperature probes.
The 9103 covers below-ambient temperatures as low as -25°C. The 9103 is stable to ±0.02°C, and its display is calibrated to an accuracy of ±0.25°C at all temperatures within its range. In just eight minutes, 0°C is reached, and 100°C is reached in six minutes, so your time is spent calibrating-not waiting.
The 9103 Dry-well was designed for portability, low cost, and ease of operation. Through proper use, the instrument will continuously provide accurate calibration of temperature sensors and devices.
Lightweight and very portable.
Accuracy to ± 0.25 °C.
RS-232 and Interface-it software included.
Easy to recalibrate.
|Range||–25°C to 140°C (–13°F to 284°F) at 23°C ambient.|
|Accuracy||±0.25°C (holes greater than 1/4" [6.35mm]: ±1°C).|
|Stability||±0.02°C at –25°C
±0.04°C at 140°C.
|Well-to-well uniformity||±0.1°C between similarly sized wells.|
|Heating times||18 minutes from ambient to 140°C.|
|Cooling times||20 minutes from ambient to –25°C.|
|Stabilisation time||7 minutes.|
|Immersion depth||124 mm (4.875 in).|
|Inserts||Insert A, B, C, or D included (specify when ordering).|
|Outside insert dimensions||31.8 mm dia. x 124 mm (1.25 x 4.88 in).|
|Computer interface||RS-232 included with free Interface-it software (Model 9930).|
|Power||115 V AC (±10%), 1.3 A or 230 V AC (±10%), 0.7 A, switchable, 50/60 Hz, 150 W.|
|Size (W x H x D)||143 x 261 x 245 mm
(5.63 x 10.25 x 9.63 in).
|Weight||5.7 kg (12 lb).|
|NIST-traceable certificate||Data at –25°C, 0°C, 25°C, 50°C, 75°C, 100°C, and 140°.|
No customer reviews for the moment.